This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||8 August 2011|
|PDF File Size:||6.52 Mb|
|ePub File Size:||9.69 Mb|
|Price:||Free* [*Free Regsitration Required]|
AO required Adjustable range and aairanger limits. When the reported level reaches Unless altered, the sound velocity used for the distance calculation is Enter the critical point for the desired action to occur based on the Relay Function selected.
Contact the seller – opens in a new window or tab and request post to your location. Increase Measurement Airanfer as required to stabilize the Reading, or install a stilling well contact Milltronics or your local distributor. A simple calibration feature compensates for homogeneous consistent composition atmospheres other than air. Page 68 Extend Near Blanking to overcome measurement difficulties which cannot be corrected by transducer location, mounting, or aiming.
On successful test completion, “PASS” is displayed. Comments to this Manuals Your Name.
Milltronics AiRanger XPL Plus Instruction Manual
Page 44 Allocation P range. Press to scroll access the failsafe options. For aoranger information, see the Global Shipping Programme terms and conditions – opens in a new window or tab. Enter the delay in seconds before the display advances to the next Point Number. Page Cable Transducer: Key in the desired level or volume value. However, it can be reduced under certain circum- stances. Use TS-3 temperature sensors only.
Not all components or their maximum quantity may be required. An XPL Plus level monitor, utilizing a single Siemens Milltronics ultrasonic transducer, accurately monitors vessel material level without material contact. Distance Measurement less than 60 m.
Milltronics airanger XPL plus Instruction Manual
P Span Enter “3” for 3 ft vessel diameter. Comments to this Manuals Your Name. If the material presents a flat surface perpendicular to the transducer beam, select liquid. Enhancement Parameters Scope Displays, P The register is reset to 0 by master reset P Use these features to record and save a total of up to 10 Echo profiles, initiated manually Por automatically P et al.
Enter a negative value if Empty is below sea level. Add to basket. Keypad Parameter Value in memory complete a parameter reset. Page P to stabilize the “shot” mode used for the echo evaluation. Pluw the vessel to be monitored does not match any of the 8 preset Tank Shape calculations, a Universal Volume calculation may be used.
Page 4 Measurement Parameters P to P The xiranger level is a combination of transient acoustic noise and electrical noise induced into the transducer cable or receiving circuitry itself. If Temp Source P is set to any value other than Airznger Temp, the value displayed aianger the temperature measured. Immune to power interruptions, all programming is stored indefinitely. Instruction Manual March If the material monitored is a liquid, check for splashing in the vessel. The seller won’t accept returns for this item.
Enter a value slightly greater than the maximum vessel filling rate. Place the infrared programmer in the enclosure lid recess no wiring or fastening required. Don’t have an account? Take a look at our Returning an item help page for airangre details.
Siemens SITRANS LU02 Dual Point Ultrasonic Level Transmitter (Milltronics AiRanger DPL Plus)
Got it, continue to print. P Units Enter “4” Page 71 Enter llus width of false echoes in ms to be removed from the long shot Echo Profile. Avoid entering a value that, when multiplied by the maximum current reading, could exceed 5 digits before the Decimal Position values: Not all components or their maximum quantity may be required. Smartlinx Parameters This register increments by 1 each time a bus error P is reported.
If the material monitored is a liquid, airajger for splashing in the vessel. P value P preset value 1, max verification