The compactplus integrates the latest technological advances whilst maintaining simplicity and reliability through maintaining the proven robustness of the. Service Manual Perfusor® compact, german . B. Braun also recommends training on the Technical Safety. Checks, or to perform at least . Perfusor® compact S satisfies all safety stan- dards for medical electrical .. Syringe Type. B. Braun. Code No1). Cat. No.2). Min. Vol.3). [ml]. Max. Rate4). [ml/ h].
|Published (Last):||26 September 2012|
|PDF File Size:||13.78 Mb|
|ePub File Size:||18.83 Mb|
|Price:||Free* [*Free Regsitration Required]|
Note Do not squeeze motor cable. Checks after Repair Operation 1.
B I BRAUN Perfusor COMPACT
Is the MFC correctly connected? Page 56 Unit Elements For your notes: The piston rod symbol must flash on the LC display if a syringe was not inserted.
The unit has the CE label.
Wait until the current step gauge is completely released. Battery Compartment Cover Unit Elements 4.
Small parts kit for 5 units Insert syringe and confirm or selecte. The standard delivery rate range is 0.
Unit Elements this straight pin lock is already fitted. Page 54 Unit Elements 3. Note The following staff call modes can be selected in the Service Program if the unit is switched peffusor B BraunPumps Tags: The individual steps are described hereafter in more detail.
Perfusor® compact plus
Syringe holder, axial positioner, drive head Syringe table and quick reference guide Membrane keyboard Battery compartment cover, battery compartment and – contacts Unit feet MFC connector Changed manual structure New software To print the manual completely, please, download it. New type plates can only be ordered as spare parts if the old type plates are returned to B.
The error code of the control microprocessor can be queried with the F button. Screw out screwed split rivet. Is syringe selection set to “free type”, but “free type” Open housing carefully, then 3. The pumps slide and lock onto one another to form a multi-channel device with PCA, large volume and syringe pump capabilities.
Fundamental Repair Information If the battery pack is used, then the device is to be switched on with line connection. You will be included in the revision service after: N-Module V Open lock drive head. Unit Elements tables differ provide the unit with the new table supplied with the E-Module.
Used B BRAUN Perfusor Compact Pump IV Infusion For Sale – DOTmed Listing #
A-module Unit Elements 4. The easy-to-learn and easy-to-program system includes built-in error reduction software and a medication library accommodating both hard and soft limits, large navigational arrows and compsct screen that remains visible in any light condition. System Overview Accessories Designation Ord.
Got it, continue to print. Perfisor head housing Cap for drive head housing 4. Note Defective batteries must be disposed of according to the regulations, e. Page of 86 Go. Open syringe holder, an alarm is triggered.